@article {2783, title = {Axial CID and high pressure resonance CID in a miniature ion trap mass spectrometer using a discontinuous atmospheric pressure interface.}, journal = {Journal of the American Society for Mass Spectrometry}, volume = {21}, year = {2010}, month = {2010 Feb}, pages = {209-14}, abstract = {

Axial collision induced dissociation (CID) and high-pressure resonance CID were implemented and compared with normal low-pressure resonance CID in a miniature ion trap mass spectrometer to obtain more complete fragmentation spectra. Axial CID was realized simply by applying a potential to the discontinuous atmospheric pressure interface (DAPI) capillary without performing parent ion isolation before dissociation. High-pressure resonance CID employed a double-introduction pulse scan function, by means of which precursor ions isolated at low-pressure (\<10(-3) torr) were dissociated at high-pressure (0.1 torr-1 torr) with higher excitation energy, so that tandem MS of isolated precursor ions was achieved and extensive fragmentation was obtained. A simple peptide (Leu-enkephalin) and dye molecule (rhodamine B) ionized by ESI were used to investigate both methods and compare them with normal low-pressure resonance CID.

}, issn = {1879-1123}, doi = {10.1364/AO.50.001792}, author = {Gao, Liang and Li, Guangtao and Cooks, R Graham} } @article {2784, title = {Direct detection of benzene, toluene, and ethylbenzene at trace levels in ambient air by atmospheric pressure chemical ionization using a handheld mass spectrometer.}, journal = {Journal of the American Society for Mass Spectrometry}, volume = {21}, year = {2010}, month = {2010 Jan}, pages = {132-5}, abstract = {

The capabilities of a portable mass spectrometer for real-time monitoring of trace levels of benzene, toluene, and ethylbenzene in air are illustrated. An atmospheric pressure interface was built to implement atmospheric pressure chemical ionization for direct analysis of gas-phase samples on a previously described miniature mass spectrometer (Gao et al. Anal. Chem.2006, 78, 5994-6002). Linear dynamic ranges, limits of detection and other analytical figures of merit were evaluated: for benzene, a limit of detection of 0.2 parts-per-billion was achieved for air samples without any sample preconcentration. The corresponding limits of detection for toluene and ethylbenzene were 0.5 parts-per-billion and 0.7 parts-per-billion, respectively. These detection limits are well below the compounds{\textquoteright} permissible exposure levels, even in the presence of added complex mixtures of organics at levels exceeding the parts-per-million level. The linear dynamic ranges of benzene, toluene, and ethylbenzene are limited to approximately two orders of magnitude by saturation of the detection electronics.

}, keywords = {Air, Atmospheric Pressure, Benzene, Benzene Derivatives, Equipment Design, Ions, Limit of Detection, Mass Spectrometry, Toluene}, issn = {1879-1123}, doi = {10.1364/AO.50.001792}, author = {Huang, Guangming and Gao, Liang and Duncan, Jason and Harper, Jason D and Sanders, Nathaniel L and Ouyang, Zheng and Cooks, R Graham} } @article {2785, title = {Ion trap mass analysis at high pressure: an experimental characterization.}, journal = {Journal of Mass Spectrometry}, volume = {45}, year = {2010}, month = {2010 Jan}, pages = {26-34}, abstract = {

In recent years, it has become increasingly interesting to understand the performance of mass spectrometers at pressures much higher than those employed with conventional operating conditions. This interest has been driven by several influences, including demand for the development of reduced-power miniature mass spectrometers, desire for improved ion transfer into and through mass spectrometers, enhanced-yield preparative mass separations, and mass filtering at the atmospheric pressure interface. In this study, an instrument was configured to allow for the performance characterization of a rectilinear ion trap (RIT) at pressures up to 50 mtorr with air used as the buffer gas. The mass analysis efficiency, mass resolution, isolation efficiency, and collision-induced dissociation (CID) efficiency were evaluated at pressures ranging from 1 to 50 mtorr. The extent of degradation of mass resolution, isolation efficiency and ion stability as functions of pressure were characterized. Also, the optimal resonance ejection conditions were obtained at various pressures. Operations at 50 mtorr demonstrated improved CID efficiency in addition to peak widths of 2 and 5 m/z units (full width at half-maximum, FWHM) for protonated caffeine (m/z 195) and Ultramark (m/z 1521) respectively.

}, keywords = {Equipment Design, Ions, Mass Spectrometry, Pressure, Proteins}, issn = {1096-9888}, doi = {10.1364/AO.50.001792}, author = {Song, Qingyu and Xu, Wei and Smith, Scott A and Gao, Liang and Chappell, William J and Cooks, R Graham and Ouyang, Zheng} } @article {2796, title = {In situ SIMS analysis and reactions of surfaces prepared by soft landing of mass-selected cations and anions using an ion trap mass spectrometer.}, journal = {Journal of the American Society for Mass Spectrometry}, volume = {20}, year = {2009}, month = {2009 Jun}, pages = {949-56}, abstract = {

Mass-selected polyatomic cations and anions, produced by electrosonic spray ionization (ESSI), were deposited onto polycrystalline Au or fluorinated self-assembled monolayer (FSAM) surfaces by soft landing (SL), using a rectilinear ion trap (RIT) mass spectrometer. Protonated and deprotonated molecules, as well as intact cations and anions generated from such molecules as peptides, inorganic catalysts, and fluorescent dyes, were soft-landed onto the surfaces. Analysis of the modified surfaces was performed in situ by Cs(+) secondary ion mass spectrometry (SIMS) using the same RIT mass analyzer to characterize the sputtered ions as that used to mass select the primary ions for SL. Soft-landing times as short as 30 s provided surfaces that yielded good quality SIMS spectra. Chemical reactions of the surfaces modified by SL were generated in an attached reaction chamber into which the surface was transferred under vacuum. For example, a surface on which protonated triethanolamine had been soft landed was silylated using vapor-phase chlorotrimethylsilane before being returned still under vacuum to the preparation chamber where SIMS analysis revealed the silyloxy functionalization. SL and vapor-phase reactions are complementary methods of surface modification and in situ surface analysis by SIMS is a simple way to characterize the products produced by either technique.

}, issn = {1879-1123}, doi = {10.1364/AO.50.001792}, author = {Nie, Zongxiu and Li, Guangtao and Goodwin, Michael P and Gao, Liang and Cyriac, Jobin and Cooks, R Graham} } @article {2798, title = {Mass selection of ions from beams using waveform isolation in radiofrequency quadrupoles.}, journal = {Analytical Chemistry}, volume = {81}, year = {2009}, month = {2009 Mar 1}, pages = {1833-40}, abstract = {

A waveform isolation method is described for the mass-selective transmission of ions through quadrupole mass filters, and it is implemented on a new tandem mass analyzer instrument. The method features the application of broad-band waveforms comprising appropriate frequencies to cause mass-selective instability in ions of particular mass-to-charge (m/z) and to transmit all others. The experiment is implemented in a tandem quadrupole system in which the first mass filter is a rectilinear ion trap (RIT) operated in a continuous mass-selective mode to transmit ions of ions of one or more arbitrarily selected m/z value(s). The second analyzer was used to verify the quality of the mass selection achieved using the first analyzer via conventional quadrupole ion trap mass-selective instability scanning. A new subtype of product ion tandem mass spectrometry (MS/MS) scan, termed the summed product ion scan, is demonstrated with a mixture of biological compounds. It is used to characterize product ions arising after simultaneous isolation and collisional activation of multiple precursor species, in this case ions of the same analyte generated in different charge states. The summed product ion scan can be useful for enhancing sensitivity for the analyte of interest or for providing more comprehensive information about an analyte than is available by monitoring a single ionized form of the analyte. The analytical performance of the waveform isolation method is tested using simple drug mixtures, and its potential for increasing overall yields in preparative mass spectrometry is explored briefly. It is shown that efficiencies of ca. 70\% of ion transfer to a surface for ion soft landing surface can be achieved. The upper mass range is limited by axial acceleration arising from the stretched geometry, and one solution to this problem is provided.

}, keywords = {Carmustine, Cytarabine, Electrospray Ionization, Ions, Mass, Mass Spectrometry, Mathematics, Melphalan, Radio Waves, Spectrometry, Tandem Mass Spectrometry}, issn = {1520-6882}, doi = {10.1364/AO.50.001792}, author = {Song, Qingyu and Smith, Scott A and Gao, Liang and Xu, Wei and Voln{\'y}, Michael and Ouyang, Zheng and Cooks, R Graham} } @article {2803, title = {Breaking the pumping speed barrier in mass spectrometry: discontinuous atmospheric pressure interface.}, journal = {Analytical Chemistry}, volume = {80}, year = {2008}, month = {2008 Jun 1}, pages = {4026-32}, abstract = {

The performance of mass spectrometers with limited pumping capacity is shown to be improved through use of a discontinuous atmospheric pressure interface (DAPI). A proof-of-concept DAPI interface was designed and characterized using a miniature rectilinear ion trap mass spectrometer. The interface consists of a simple capillary directly connecting the atmospheric pressure ion source to the vacuum mass analyzer region; it has no ion optical elements and no differential pumping stages. Gases carrying ionized analytes were pulsed into the mass analyzer for short periods at high flow rates rather than being continuously introduced at lower flow rates; this procedure maximized ion transfer. The use of DAPI provides a simple solution to the problem of coupling an atmospheric pressure ionization source to a miniature instrument with limited pumping capacity. Data were recorded using various atmospheric pressure ionization sources, including electrospray ionization (ESI), nano-ESI, atmospheric pressure chemical ionization (APCI), and desorption electrospray ionization (DESI) sources. The interface was opened briefly for ion introduction during each scan. With the use of the 18 W pumping system of the Mini 10, limits of detection in the low part-per-billion levels were achieved and unit resolution mass spectra were recorded.

}, keywords = {Air, Atmospheric Pressure, Electrospray Ionization, Gases, Laboratories, Mass, Mass Spectrometry, Spectrometry, Tandem Mass Spectrometry, Time Factors}, issn = {1520-6882}, doi = {10.1364/AO.50.001792}, author = {Gao, Liang and Cooks, R Graham and Ouyang, Zheng} } @article {2802, title = {Design and characterization of a multisource hand-held tandem mass spectrometer.}, journal = {Analytical Chemistry}, volume = {80}, year = {2008}, month = {2008 Oct 1}, pages = {7198-205}, abstract = {

A wireless-controlled miniature rectilinear ion trap mass spectrometer system, total weight with batteries 5.0 kg, consuming less than 35 W of power, and having dimensions of 22 cm in length by 12 cm in width by 18 cm in height, is characterized. The design and construction of the mass spectrometer including mass analyzer, vacuum system, electronics system, and data acquisition and processing systems, is detailed. The mass spectrometer is compatible with various types of ionization sources including a glow discharge electron impact ionization source used in the internal ionization mode, and various atmospheric pressure ionization sources, including electrospray ionization, atmospheric pressure chemical ionization, and desorption electrospray ionization, which are employed for external, atmospheric pressure ionization. These external sources are coupled to the miniature mass spectrometer via a capillary interface that is operated in a discontinuous fashion (discontinuous atmospheric pressure interface) to maximize ion transport. The performance of the mass spectrometer for large and small molecules is characterized. Limits of detection in the parts-per-billion range were obtained for selected compounds examined using both the internal ionization and external ionization modes. Tandem mass spectrometry and fast in situ analysis capabilities are also demonstrated using a variety of compounds and ionization sources. Protein molecules are analyzed as the multiply protonated molecules with mass/charge ratios up to 1500 Da/charge.

}, keywords = {Miniaturization, Sensitivity and Specificity, Tandem Mass Spectrometry}, issn = {1520-6882}, doi = {10.1364/AO.50.001792}, author = {Gao, Liang and Sugiarto, Andy and Harper, Jason D and Cooks, R Graham and Ouyang, Zheng} } @article {2806, title = {Monitoring of toxic compounds in air using a handheld rectilinear ion trap mass spectrometer.}, journal = {Analytical Chemistry}, volume = {80}, year = {2008}, month = {2008 Feb 1}, pages = {734-41}, abstract = {

A miniature, handheld mass spectrometer, based on the rectilinear ion trap mass analyzer, has been applied to air monitoring for traces of toxic compounds. The instrument is battery-operated, hand-portable, and rugged. We anticipate its use in public safety, industrial hygiene, and environmental monitoring. Gaseous samples of nine toxic industrial compounds, phosgene, ethylene oxide, sulfur dioxide, acrylonitrile, cyanogen chloride, hydrogen cyanide, acrolein, formaldehyde, and ethyl parathion, were tested. A sorption trap inlet was constructed to serve as the interface between atmosphere and the vacuum chamber of the mass spectrometer. After selective collection of analytes on the sorbent bed, the sorbent tube was evacuated and then heated to desorb analyte into the instrument. Sampling, detection, identification, and quantitation of all compounds were readily achieved in times of less than 2 min, with detection limits ranging from 800 parts per trillion to 3 parts per million depending on the analyte. For all but one analyte, detection limits were well below (3.5-130 times below) permissible exposure limits. A linear dynamic range of 1-2 orders of magnitude was obtained over the concentration ranges studied (sub-ppbv to ppmv) for all analytes.

}, issn = {0003-2700}, doi = {10.1364/AO.50.001792}, author = {Keil, Adam and Hernandez-Soto, Heriberto and Noll, Robert J and Fico, Miriam and Gao, Liang and Ouyang, Zheng and Cooks, R Graham} } @article {2810, title = {Ambient mass spectrometry with a handheld mass spectrometer at high pressure.}, journal = {Analytical Chemistry}, volume = {79}, year = {2007}, month = {2007 Oct 15}, pages = {7734-9}, abstract = {

The first coupling of atmospheric pressure ionization methods, electrospray ionization (ESI) and desorption electrospray ionization (DESI), to a miniature hand-held mass spectrometer is reported. The instrument employs a rectilinear ion trap (RIT) mass analyzer and is battery-operated, hand-portable, and rugged (total system: 10 kg, 0.014 m(3), 75 W power consumption). The mass spectrometer was fitted with an atmospheric inlet, consisting of a 10 cm x 127 microm inner diameter stainless steel capillary tube which was used to introduce gas into the vacuum chamber at 13 mL/min. The operating pressure was 15 mTorr. Ions, generated by the atmospheric pressure ion source, were directed by the inlet along the axis of the ion trap, entering through an aperture in the dc-biased end plate, which was also operated as an ion gate. ESI and DESI sources were used to generate ions; ESI-MS analysis of an aqueous mixture of drugs yielded detection limits in the low parts-per-billion range. Signal response was linear over more than 3 orders of magnitude. Tandem mass spectrometry experiments were used to identify components of this mixture. ESI was also applied to the analysis of peptides and in this case multiply charged species were observed for compounds of molecular weight up to 1200 Da. Cocaine samples deposited or already present on different surfaces, including currency, were rapidly analyzed in situ by DESI. A geometry-independent version of the DESI ion source was also coupled to the miniature mass spectrometer. These results demonstrate that atmospheric pressure ionization can be implemented on simple portable mass spectrometry systems.

}, keywords = {Alkylation, Calibration, Mass Spectrometry, Molecular Structure, Molecular Weight, Pressure, Proteins, Temperature}, issn = {0003-2700}, doi = {10.1364/AO.50.001792}, author = {Keil, Adam and Talaty, Nari and Janfelt, Christian and Noll, Robert J and Gao, Liang and Ouyang, Zheng and Cooks, R Graham} }